|Minimum Division:||0.1% Brix||Length:||200mm|
sugar brix refractometer,
brix scale refractometer
Yieryi 0-10 Brix Low Sugar Fruits And Vegetables Juices Portable Brix refractometer
The series is developed for working with sugar related liquids (fruit juices, soft drinks, wine), help monitor
and control sugar concentrations in foods and beverages. Whether users are checking the "ripeness" of
fruit in the field, verifying product quality after harvesting, or controlling concentrations during processing
and packaging, refractometers provide critical information to ensure product quality. But,it is also
commonly used for controlling the concentration of various industrial fluids cutting lubricants and flux
rinsing compounds). Full ranges of measurement is available.
1. Measuring Range: 0-10% Brix
2. Minimum Division: 0.1% Brix
3. Accuracy: ±0.10%
4. ATC Compensation Range: 10°C~30°C (50°F~86°F)
5. Manual calibration with mini-screw driver provided
6. Length: 200mm
7. Weight: 165g
8. Durable and built to last long**
9. Uses ambient light only which means battery or power source is not required*
10. Heavy-duty and lightweight with its aluminum construction*
11. Easy to focus and calibrate*
12. Accurate testing results guaranteed*
13. Made with the highest and finest quality of aluminum & rubber that makes it lightweight*
14. Cushioned with soft & comfortable non-slip rubber*
15. Extremely easy-to-use and calibrate
Precautions & Maintenance
After using, please do not use the tap water to clean directly; Otherwise the water may go into the pipe of
When using or maintaining, please handle with care and keep the surface of optical parts away from
scratch or demage.
The instrument can only be stored in the environment where is dry, free of dirt and corroisive gas;
otherwise the optical part will get moundly.
The prism that will be contacted with the object under measurement is an easily –consumed product,
which can not be recovered, pls understand.
Users are required to follow the methods said above; If the instruments is demaged since misconduct or
operation in error, we shall not bear the warranty responsibility.
The instrument will be packaged into the special box, with the accessories (i.e.:Using Instruction For Use;
Screwdriver, Suction pipe and lint)
Contact Person: Sunny