|Brix Scale:||0 To 10%Brix||Reference Temperature:||20°C|
|Compensation Range:||10°C~30°C (50°F~86°F)||SKU:||RHS-10ATC|
|Name:||Digital Salinity Refractometer||Feature:||Durable|
Glycol Tester Refractometer,
Digital Honey Refractometer,
ATC Digital Salinity Refractometer
Yieryi brix 0-10% refractometer 1.000-1.070SG Salinity Seawater salinometer ATC 10 to 30C
Precautions & Maintenance
After using, please do not use the tap water to clean directly; Otherwise the water may go into the pipe of
When using or maintaining, please handle with care and keep the surface of optical parts away from
scratch or demage.
The instrument can only be stored in the environment where is dry, free of dirt and corroisive gas;
otherwise the optical part will get moundly.
The prism that will be contacted with the object under measurement is an easily –consumed product,
which can not be recovered, pls understand.
Users are required to follow the methods said above; If the instruments is demaged since misconduct or
operation in error, we shall not bear the warranty responsibility.
The instrument will be packaged into the special box, with the accessories (i.e.:Using Instruction For Use;
Screwdriver, Suction pipe and lint)
1. Let the deviation prism point to the light, adjust the visual ring till the gage mark becomes clear.
2. Adjustment benchmark: Before using, pls keep the standard liquor (purified water), instrument and
liquid under measurement at the same temperature.Open the cover plate and apply 1~2 drops of
standard liquor onto the diviation prism, and then use hand to press the cover plate lightly to get a
dividing line. Turn the calibration screw to coincide the dividing line with the benchmark (0%). The one for
honey will be an exception.
Note: The instrument has been calibrated before delivering, you are allowed to use it directly.
3. Open the cover plate and use the soft rag to clear the surface of prism: apply 1~2 drops of liquid under
measurement onto the deviation prism and close the cover plate to press lightly; afterwards read the
relevant scale for dividing line; the reading will be the concentration value and destiny value of the liquid
4. After measuring, use the wet rag to clean the prism and cover plate; when it becomes dry, please store
the prism safely.
Contact Person: Sunny